- Quantitative, non-subjective results
- Increased sensitivity and broader detection range compared to rapid tests
Working range: 0.30-80.00 ng/mL
Intra-lot precision was determined by testing of PROG reference materials using 10 test devices from the same lot. CV ≤ 15%.
Inter-lot precision was determined by testing of PROG reference materials using 30 test devices from 3 consecutive lots randomly (10 test devices from each lot). CV ≤ 20%.
PROG control materials with two different concentrations were tested by every lot of test device, and the deviations were within ± 15%.
A serial concentration of PROG reference materials at 3.00-80.00 ng/mL was tested, and the correlation coefficient (R) is ≥ 0.9900.
|Crossed Factors||Concentration||Test Result|
|Estradiol||500 ng/mL||< 0.3 ng/mL|
|Testosterone||500 ng/mL||< 0.3 ng/mL|
|Cortisol||500 ng/mL||< 0.3 ng/mL|
|Aldosterone||500 ng/mL||< 0.3 ng/mL|
|Estriol||100 ng/mL||< 0.3 ng/mL|
|17-β Estradiol||100 ng/mL||< 0.3 ng/mL|
|Oestrone||100 ng/mL||< 0.3 ng/mL|
|17-α hydroxyl progesterone||100 ng/mL||< 0.3 ng/mL|